Mechanical effects of surgical procedures on osteochondral grafts elucidated by osmotic loading and real-time ultrasound
© Hattori et al.; licensee BioMed Central Ltd. 2009
Received: 19 May 2009
Accepted: 2 September 2009
Published: 2 September 2009
Osteochondral grafts have become popular for treating small, isolated and full-thickness cartilage lesions. It is recommended that a slightly oversized, rather than an exact-sized, osteochondral plug is transplanted to achieve a tight fit. Consequently, impacting forces are required to insert the osteochondral plug into the recipient site. However, it remains controversial whether these impacting forces affect the biomechanical condition of the grafted articular cartilage. The present study aimed to investigate the mechanical effects of osteochondral plug implantation using osmotic loading and real-time ultrasound.
A full-thickness cylindrical osteochondral defect (diameter, 3.5 mm; depth, 5 mm) was created in the lateral lower quarter of the patella. Using graft-harvesting instruments, an osteochondral plug (diameter, 3.5 mm as exact-size or 4.5 mm as oversize; depth, 5 mm) was harvested from the lateral upper quarter of the patella and transplanted into the defect. Intact patella was used as a control. The samples were monitored by real-time ultrasound during sequential changes of the bathing solution from 0.15 M to 2 M saline (shrinkage phase) and back to 0.15 M saline (swelling phase). For cartilage sample assessment, three indices were selected, namely the change in amplitude from the cartilage surface (amplitude recovery rate: ARR) and the maximum echo shifts from the cartilage surface and the cartilage-bone interface.
The ARR is closely related to the cartilage surface integrity, while the echo shifts from the cartilage surface and the cartilage-bone interface are closely related to tissue deformation and NaCl diffusion, respectively. The ARR values of the oversized plugs were significantly lower than those of the control and exact-sized plugs. Regarding the maximum echo shifts from the cartilage surface and the cartilage-bone interface, no significant differences were observed among the three groups.
These findings demonstrated that osmotic loading and real-time ultrasound were able to assess the mechanical condition of cartilage plugs after osteochondral grafting. In particular, the ARR was able to detect damage to the superficial collagen network in a non-destructive manner. Therefore, osmotic loading and real-time ultrasound are promising as minimally invasive methods for evaluating cartilage damage in the superficial zone after trauma or impact loading for osteochondral grafting.
Osteochondral grafts have become popular for the treatment of small, isolated and full-thickness cartilage lesions . Osteochondral grafts have several advantages, including a high survival rate of the grafted articular cartilage, reliable bone union and no threat of disease transmission [1–3]. Several osteochondral transplantation systems are commercially available in clinical practice. For most of these systems, it is recommended that a slightly oversized, rather than an exact-sized, osteochondral plug is transplanted to achieve a tight fit , because plug stability is an important factor for optimal in-growth of a transplanted plug . Therefore, impacting forces are required to insert the osteochondral plug into the recipient site during the osteochondral grafting procedure.
It remains controversial whether the impacting forces required to insert an osteochondral plug affect the biomechanical condition of the grafted articular cartilage. We previously developed an ultrasonic evaluation system for articular cartilage. We demonstrated that this system can be used to quantitatively clinically evaluate cartilage degeneration [6, 7]. Using the same ultrasonic evaluation system, Kuroki and colleagues  examined the mechanical effects of the osteochondral grafting procedure on porcine articular cartilage immediately after surgery. The study indicated that osteochondral graft surgery does not affect the stiffness, surface irregularity or thickness of either oversized and exact-sized plugs. In contrast, Nishitani and colleagues  assessed osteochondral grafting of the human elbow using this system and showed that the cartilage plug may become damaged during the osteochondral grafting procedure. Nakaji and colleagues  evaluated the mechanical properties of cartilage plugs using a tactile sensor system and showed that the stiffness of oversized cartilage plugs did not differ significantly from that of the normal cartilage immediately after surgery. However, it is well known that the impacting forces required to implant an osteochondral graft can lead to chondrocyte death and fissure formation in the surface of the cartilage plug [11, 12]. Therefore, it is speculated that the above described evaluation methods are not suitable for the assessment of articular cartilage damage from the impacting forces used to implant an osteochondral graft. Therefore, a more adjustable measurement method is required.
Ultrasound was first used to measure the osmotic swelling of articular cartilage by Tepic and colleagues . Further studies have recently been carried out by Zheng and colleagues  and Wang and colleagues [15, 16], who developed a new ultrasound system for monitoring transient depth-dependent osmotic swelling and solute diffusion in articular cartilage. Using this system, they successfully monitored articular cartilage digestion by trypsin in real time. Ultrasound assessment by osmotic loading can provide transient and depth-dependent swelling information for articular cartilage in situ. Therefore, osmotic loading and real-time ultrasound have the potential for assessing the cartilage damage caused by the impacting forces required to insert a plug during the osteochondral graft procedure. However, it remains unknown whether osmotic loading and real-time ultrasound can assess the mechanical condition of a cartilage plug after osteochondral grafting.
The purpose of the present study was to evaluate the mechanical effects of osteochondral plug implantation using osmotic loading and real-time ultrasound and to demonstrate the accuracy of ultrasound in identifying the cartilage damage after osteochondral graft procedures. To this end, we evaluated oversized and exact-sized cartilage plugs after osteochondral grafting. In the present study, we also assessed the cartilage plugs using a conventional mechanical test and observed the cartilage surface morphology by scanning electron microscopy (SEM).
Materials and methods
Cartilage sample processing
Porcine knee joints (n = 30) with intact capsules and ligaments were purchased from a slaughterhouse. After removal of the soft tissues, the knee joints were opened. The patellas with visually intact surfaces were harvested, wrapped in wet gauze soaked with physiological saline solution and stored at -20°C until use. For sample preparation, each patella was thawed at room temperature for one hour and immersed in physiological saline solution (0.15 M sodium chloride (NaCl)), before the lateral lower and upper quarters of the patella were cut using a band saw (K-100; Hozan Tool Industrial Co. Ltd., Osaka, Japan). During the processing steps described below, the cartilage surface was kept moist with physiological saline solution without immersing the sample.
Ultrasound monitoring system
where MAMP swelling is the mean amplitude from the cartilage surface in the swelling phase, and MAMP shrinkage is the mean amplitude from the cartilage surface in the shrinkage phase.
We also evaluated the echo shifts from the cartilage surface and the cartilage-bone interface in both the shrinkage and swelling phases. The echo shift from the cartilage surface indicates the sample displacement, while the echo shift from the cartilage-bone interface indicates the diffusivity of the saline solution in the sample . Therefore, as the other quantitative indices of the cartilage assessment in this study, the maximum echo shifts were chosen.
Two samples in each group were subjected to morphological analysis using an SEM (Model SM-350; Topcon Technohouse Corporation, Tokyo, Japan). The samples were fixed in 2% glutaraldehyde buffered with 0.1 M cacodylate, dehydrated in a graded ethanol series, dried using the critical point technique and coated by sputtering with a gold layer .
Eight cartilage samples were immersed in physiological saline and tested within three hours. To determine the mechanical properties of the grafted cartilage, an electromechanical material testing machine (EZ-L; Shimadzu Corporation, Kyoto, Japan) was used. Forces were applied to the grafted cartilage at a displacement rate of 2.0 mm/min using a 3.0 mm diameter solid aluminum indenter. A load-deformation curve was obtained during the compression. As biomechanical parameters, we defined the maximum load (breaking load: F max) applied at fracture of the grafted cartilage.
For multiple comparisons of ultrasound findings, the groups were analyzed using the nonparametric Kruskal-Wallis test. When significant variance was detected, the differences among individual groups were determined using the Mann-Whitney U test with the Bonferroni correction. For comparisons between two groups in the biomechanics analyses, the differences were analyzed by the nonparametric Mann-Whitney U test. The significance level was set at P < 0.05.
Echo shifts from cartilage surface and cartilage-bone interface in the shrinkage and swelling phases
(n = 10)
(n = 10)
(n = 10)
-82.6 ± 26.1 ns
-77.4 ± 22.7 ns
-70.7 ± 27.8 ns
5.2 ± 24.8 ns
14.2 ± 24.8 ns
22.4 ± 16.6 ns
-9.2 ± 21.5 ns
-2.6 ± 15.0 ns
4.4 ± 9.6 ns
-86.0 ± 18.1 ns
-74.8 ± 12.9 ns
-69.1 ± 19.2 ns
The present study investigated the osmotic shrinkage-swelling behaviours of oversized and exact-sized cartilage plugs in osteochondral grafting using osmotic loading and real-time ultrasound. The main findings of the study are that osmotic loading and real-time ultrasound are capable of assessing the mechanical condition of a cartilage plug after osteochondral grafting. In particular, the ARR was able to detect damage to the superficial collagen network in a non-destructive manner. Therefore, osmotic loading and real-time ultrasound are promising as minimally invasive methods for evaluating cartilage damage in the superficial zone after trauma or impact loading for osteochondral grafting.
An osteochondral plug that is exactly the same size and shape as a cartilage defect seems to be ideal for osteochondral grafting. However, Makino and colleagues  reported that histological changes occur in the implanted cartilage, after examining osteochondral grafts taken from the femoral condyle and returned to their original sites. In their rabbit model, the graft was not strictly the same size as the defect because of the blade thickness of the chisel used to take the graft. Moreover, they revealed that an oversized osteochondral graft appeared to be almost the same as the normal adjacent cartilage at 4, 12 and 24 weeks after surgery . Therefore, an oversized plug can be recommended for use in the osteochondral graft procedure. However, the impact load required to insert a plug into the recipient site is higher for an oversized plug than for an exact-sized plug.
Impact loading of articular cartilage has commonly been associated with structural damage [19–22], loss of viability and changes in the metabolism of chondrocytes [19, 22–24], with subsequent degeneration of the articular cartilage . In general, evaluations of damage to cartilage have been performed by histological analysis of the structural integrity [19, 22], SEM imaging of the surface morphology , assessment of tissue swelling by the water content related to disruption of collagen fibrils [19, 23], assessment of chondrocyte death [19, 24] and release of cartilage macromolecular constituents during subsequent tissue culture [19, 22, 24]. However, these analyses require the collection of cartilage tissue samples, which will result in damage to the cartilage plug surface. Therefore, all the above described evaluation methods should be avoided in clinical practice.
There are several imaging modalities to assess articular cartilage such as radiograph, computed tomography (CT), magnetic resonance imaging (MRI) and optical coherence tomography (OCT). Radiograph and CT do not image soft tissue, which prevent identification of structural changes of articular cartilage. Conventional MRI has been used in clinical practice to measure morphological change in articular cartilage. In comparison with MRI, the present ultrasonic approach may allow real-time monitoring of depth-dependent osmotic behaviours by the echo shift and the changes in amplitude. Moreover, the present system is much less expensive in comparison with MRI. OCT is a novel form of optical imaging that enables cross-sectional visualization of tissue micro architecture. However, OCT is still in its early stages of development for the assessment of articular cartilage [26, 27]. Therefore, further studies to assess articular cartilage from the view point of biomechanics are required.
Tepic and colleagues  developed an ultrasonic system for assessing osmotic swelling of articular cartilage after dehydration in humid air. However, their ultrasonic system was only able to evaluate the whole cartilage layer and no measurements were obtained for depth-dependent swelling behaviours. Zheng and colleagues developed a new ultrasound system for monitoring transient depth-dependent osmotic swelling and solute diffusion in articular cartilage [14–16]. Consequently, osmotic loading and real-time ultrasound can provide comprehensive information about the biomechanical behaviour of articular cartilage. The present study has demonstrated the feasibility of this system for evaluating cartilage damage caused by impact loading while inserting a plug during the osteochondral graft procedure.
In this study, cartilage plugs were assessed not only by their osmotic shrinking and swelling behaviours but also by the changes in amplitude of the cartilage surface from the shrinkage phase to the swelling phase. A previous study revealed that the amplitude from the cartilage surface is related to the tissue reflection coefficient, acoustic impedance, elastic modulus and surface condition in physics, and related to proteoglycan depletion and collagen disruption in biology [28–30]. In the present study, the cartilage plugs were damaged by the impact loading required for their insertion into the defects. Moreover, damage to the surface collagen network was confirmed by SEM. By using osmotic swelling, differences in the cartilage surface integrity between oversized cartilage plugs and intact cartilage were enhanced. As a result, the ARR of oversized cartilage plugs was significantly lower than that of intact cartilage. Therefore, the ARR mainly reveals the microstructural changes to the articular cartilage in the superficial collagen-rich zone.
On the other hand, the echo shift from the cartilage surface is known to reflect the sample displacement and the echo shift from the cartilage-bone interface is known to reflect the diffusivity of saline solution in the sample . In the present study, the echo shifts of oversized and exact-sized cartilage plugs were similar to those of intact cartilage. These results suggest that the interiors of the cartilage plugs were not damaged by the impact loading required to insert the plugs into the defects. Within the limitations of the measurement accuracy, the mechanical indentation test could not detect damage to the cartilage surface. Therefore, osmotic loading and real-time ultrasound represent new approaches for studying the biomechanical and biophysical aspects associated with articular cartilage.
Three limitations of our study should be considered. First, we did not examine the effects of osmotic loading on the viability and metabolism of chondrocytes. A high concentration of NaCl may be harmful to cartilage tissues. If this proves to be the case, the methodology for the osmotic loading should be changed from 2 M and 0.15 M NaCl to humid air and 0.15 M NaCl . Second, the impact loading required to insert the osteochondral plugs could not be controlled. However, the present study simulated an assessment of human osteochondral grafts, and a surgeon who was experienced in the osteochondral grafting procedure performed the harvesting and implantation procedures. Therefore, damage to the collagen network in the superficial layer of cartilage plugs would occur during the osteochondral grafting procedure.
Finally, the present study was carried out to investigate the feasibility of using osmotic loading and real-time ultrasound to assess the shrinking and swelling behaviors of cartilage plugs after osteochondral grafting. If the present study design were applied to clinical practice, the length of measurement time would come into question. However, maximum deformation of ARR and echo shift in plug cartilage by changing the saline concentration occurred during the first several minutes . Thus, with proper miniaturization of the design, it would be clinically practical to detect cartilage damage after the osteochondal graft procedure. Therefore, for application to clinical situations, further studies are required to determine whether this system will prove beneficial for the assessment of human osteochondral grafts.
The present study has obtained the first data for the assessment of articular cartilage damage caused by the impact loading required to insert an osteochondral plug using osmotic loading and real-time ultrasound. Under osmotic loading, the changes in the amplitude and echo shifts can support the evaluation of cartilage damage in osteochondral grafts. Moreover, osmotic loading and real-time ultrasound may contribute to tissue engineering in the musculoskeletal field, and the ARR and echo shifts can be expected to become quantitative indices for the biomechanical and biophysical properties of articular cartilage.
amplitude recovery rate
magnetic resonance imaging
optical coherence tomography
scanning electron microscopy.
This work was supported in part by Grants-in-Aid from the Ministry of Education, Culture, Sports, Science and Technology of Japan. The study sponsors had no role in the study design, data collection, data analysis or data interpretation, or in the writing of the report.
- Matsusue Y, Yamamuro T, Hama H: Arthroscopic multiple osteochondral transplantation to the chondral defect in the knee associated with anterior cruciate ligament disruption. Arthroscopy. 1993, 9: 318-321.View ArticlePubMedGoogle Scholar
- Hangody L, Füles P: Autologous osteochondral mosaicplasty for the treatment of full-thickness defects of weight-bearing joints. Ten years of experimental and clinical experience. J Bone Joint Surg Am. 2003, 85: 25-32.PubMedGoogle Scholar
- Hangody L, Ráthonyi GK, Duska Z, Vásárhelyi G, Füles P, Módis L: Autologous osteochondral mosaicplasty. Surgical technique. J Bone Joint Surg Am. 2004, 86: 65-72.PubMedGoogle Scholar
- Makino T, Fujioka H, Terukina M, Yoshiya S, Matsui N, Kurosaka M: The effect of graft sizing on osteochondral transplantation. Arthroscopy. 2004, 20: 837-840.View ArticlePubMedGoogle Scholar
- Makino T, Fujioka H, Yoshiya S, Terukina M, Matsui N, Kurosaka M: The effect of the small and unstable autologous osteochondral graft on repairing the full-thickness large articular cartilage defect in a rabbit model. Kobe J Med Sci. 2002, 48: 97-104.PubMedGoogle Scholar
- Hattori K, Mori K, Habata T, Takakura Y, Ikeuch K: Measurement of the mechanical condition of articular cartilage with an ultrasonic probe: quantitative evaluation using wavelet transformation. Clin Biomech (Bristol, Avon). 2003, 18: 553-557. 10.1016/S0268-0033(03)00048-2.View ArticleGoogle Scholar
- Hattori K, Takakura Y, Ishimura M, Habata T, Uematsu K, Ikeuch K: Quantitative arthroscopic ultrasound evaluation of living human cartilage. Clin Biomech (Bristol, Avon). 2004, 19: 213-216. 10.1016/j.clinbiomech.2003.11.005.View ArticleGoogle Scholar
- Kuroki H, Nakagawa Y, Mori K, Ikeuchi K, Nakamura T: Mechanical effects of autogenous osteochondral surgical grafting procedures and instrumentation on grafts of articular cartilage. Am J Sports Med. 2004, 32: 612-620. 10.1177/0095399703258744.View ArticlePubMedGoogle Scholar
- Nishitani K, Nakagawa Y, Gotoh T, Kobayashi M, Nakamura T: Intraoperative acoustic evaluation of living human cartilage of the elbow and knee during mosaicplasty for osteochondritis dissecans of the elbow: an in vivo study. Am J Sports Med. 2008, 36: 2345-2353. 10.1177/0363546508322898.View ArticlePubMedGoogle Scholar
- Nakaji N, Fujioka H, Nagura I, Kokubu T, Makino T, Sakai H, Kuroda R, Doita M, Kurosaka M: The structural properties of an osteochondral cylinder graft-recipient construct on autologous osteochondral transplantation. Arthroscopy. 2006, 22: 422-427.View ArticlePubMedGoogle Scholar
- Borazjani BH, Chen AC, Bae WC, Patil S, Sah RL, Firestein GS, Bugbee WD: Effect of impact on chondrocyte viability during insertion of human osteochondral grafts. J Bone Joint Surg Am. 2006, 88: 1934-1943. 10.2106/JBJS.E.00992.View ArticlePubMedGoogle Scholar
- Whiteside RA, Jakob RP, Wyss UP, Mainil-Varlet P: Impact loading of articular cartilage during transplantation of osteochondral autograft. J Bone Joint Surg Br. 2005, 87: 1285-1291. 10.1302/0301-620X.87B9.15710.View ArticlePubMedGoogle Scholar
- Tepic S, Macirowski T, Mann RW: Mechanical properties of articular cartilage elucidated by osmotic loading and ultrasound. Proc Natl Acad Sci USA. 1983, 80: 3331-3333. 10.1073/pnas.80.11.3331.PubMed CentralView ArticlePubMedGoogle Scholar
- Zheng YP, Shi J, Qin L, Patil SG, Mow VC, Zhou KY: Dynamic depth-dependent osmotic swelling and solute diffusion in articular cartilage monitored using real-time ultrasound. Ultrasound Med Biol. 2004, 30: 841-849. 10.1016/j.ultrasmedbio.2004.02.009.View ArticlePubMedGoogle Scholar
- Wang Q, Zheng YP, Niu HJ, Mak AF: Extraction of mechanical properties of articular cartilage from osmotic swelling behavior monitored using high frequency ultrasound. J Biomech Eng. 2007, 129: 413-422. 10.1115/1.2720919.View ArticlePubMedGoogle Scholar
- Wang Q, Zheng YP, Qin L, Huang QH, Lam WL, Leung G, Guo X, Lu HB: Real-time ultrasonic assessment of progressive proteoglycan depletion in articular cartilage. Ultrasound Med Biol. 2008, 34: 1085-1092. 10.1016/j.ultrasmedbio.2007.12.006.View ArticlePubMedGoogle Scholar
- Jurvelin J, Kuusela T, Heikkilä R, Pelttari A, Kiviranta I, Tammi M, Helminen HJ: Investigation of articular cartilage surface morphology with a semiquantitative scanning electron microscopic method. Acta Anat (Basel). 1983, 116: 302-311. 10.1159/000145755.View ArticleGoogle Scholar
- Makino T, Fujioka H, Kurosaka M, Matsui N, Yoshihara H, Tsunoda M, Mizuno K: Histologic analysis of the implanted cartilage in an exact-fit osteochondral transplantation model. Arthroscopy. 2001, 17: 747-751.View ArticlePubMedGoogle Scholar
- Jeffrey JE, Gregory DW, Aspden RM: Matrix damage and chondrocyte viability following a single impact load on articular cartilage. Arch Biochem Biophys. 1995, 322: 87-96. 10.1006/abbi.1995.1439.View ArticlePubMedGoogle Scholar
- Repo RU, Finlay JB: Survival of articular cartilage after controlled impact. J Bone Joint Surg Am. 1977, 59: 1068-1076.PubMedGoogle Scholar
- Farquhar T, Xia Y, Mann K, Bertram J, Burton-Wurster N, Jelinski L, Lust G: Swelling and fibronectin accumulation in articular cartilage explants after cyclical impact. J Orthop Res. 1996, 14: 417-423. 10.1002/jor.1100140312.View ArticlePubMedGoogle Scholar
- Ewers BJ, Dvoracek-Driksna D, Orth MW, Haut RC: The extent of matrix damage and chondrocyte death in mechanically traumatized articular cartilage explants depends on rate of loading. J Orthop Res. 2001, 19: 779-784. 10.1016/S0736-0266(01)00006-7.View ArticlePubMedGoogle Scholar
- Chen CT, Burton-Wurster N, Lust G, Bank RA, Tekoppele JM: Compositional and metabolic changes in damaged cartilage are peak-stress, stress-rate, and loading-duration dependent. J Orthop Res. 1999, 17: 870-879. 10.1002/jor.1100170612.View ArticlePubMedGoogle Scholar
- D'Lima DD, Hashimoto S, Chen PC, Colwell CW, Lotz MK: Human chondrocyte apoptosis in response to mechanical injury. Osteoarthritis Cartilage. 2001, 9: 712-719. 10.1053/joca.2001.0468.View ArticlePubMedGoogle Scholar
- Donohue JM, Buss D, Oegema TR, Thompson RC: The effects of indirect blunt trauma on adult canine articular cartilage. J Bone Joint Surg Am. 1983, 65: 948-957.PubMedGoogle Scholar
- Han CW, Chu CR, Adachi N, Usas A, Fu FH, Huard J, Pan Y: Analysis of rabbit articular cartilage repair after chondrocyte implantation using optical coherence tomography. Osteoarthritis Cartilage. 2003, 11: 111-121. 10.1053/joca.2002.0862.View ArticlePubMedGoogle Scholar
- Adams SB, Herz PR, Stamper DL, Roberts MJ, Bourquin S, Patel NA, Schneider K, Martin SD, Shortkroff S, Fujimoto JG, Brezinski ME: High-resolution imaging of progressive articular cartilage degeneration. J Orthop Res. 2006, 24: 708-715. 10.1002/jor.20083.View ArticlePubMedGoogle Scholar
- Zheng YP, Huang YP: More intrinsic parameters should be used in assessing degeneration of articular cartilage with quantitative ultrasound. Arthritis Res Ther. 2008, 10: 125-10.1186/ar2566.PubMed CentralView ArticlePubMedGoogle Scholar
- Hattori K, Ikeuchi K, Morita Y, Takakura Y: Quantitative ultrasonic assessment for detecting microscopic cartilage damage in osteoarthritis. Arthritis Res Ther. 2005, 7: R38-46. 10.1186/ar1463.PubMed CentralView ArticlePubMedGoogle Scholar
- Töyräs J, Rieppo J, Nieminen MT, Helminen HJ, Jurvelin JS: Characterization of enzymatically induced degeneration of articular cartilage using high frequency ultrasound. Phys Med Biol. 1999, 44: 2723-2733. 10.1088/0031-9155/44/11/303.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an open access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.